|Home||» Products||» Electrosurgery Plates||» Electrosurgery Plates|
REUSABLE GROUNDING PLATED FOR ELETROSURGERY
After the packaging for damages, check the plates and gel. do not use the product in the case of dames or visible defects.
Select a well - vascularised area of skin to the area to be operated. do not places the plates over scars or close to metallic prostheses or in the vicinity of ecg electrodes.
shave, degrease and dry the selected surface.
make sure that entire surface of the plates is in contact with the patients's skin.
connect the plates to the generator using the connector cable.
for connections and correct functioning, follow the instruction for use supplied by the manufacturers of the ESU unit and pencil being used.
High frequency electrosurgical generators, the grounding plates and accessories to be used with the present product should comply with the regulation in force.
The devices must be connected and put into operation only by qualified personnel.
NB: The current must never flow in a transversal direction across the body, nor cross the chest
Do not submit a patient with implanted pacemaker to electrosurgical current without first consulting a cardiologist.
For the REM type : Chech that the generator alarm systems is working before starting the procedure.
29, SahiPur Village, BH Block,
Shalimar Bagh, Delhi-110088.